|No. 8, Tianshan Road, Xinbei District, Changzhoufirstname.lastname@example.org|
|Place of Origin:||China|
|Minimum Order Quantity:||1 PCS|
|Packaging Details:||Neutral factory box|
|Delivery Time:||10-15 working days upon order quantity|
|Supply Ability:||500 pcs per month|
|Color:||As Shown In The Figure||Warranty:||2 Years|
|Delivery Time:||10-15 Working Days Upon Order Quantity||Size:||85*58*31mm|
|Name:||DC Regulated Power Supply|
dc regulated power supply,
dc dc step down power module
CNC DC regulated power supply adjustable voltage reducing module voltmeter ammeter 32V 5A160W charging
|Input voltage range:||6-40.00V|
|Output power range:||0-160W|
|Output voltage range:||0-32.00V|
|Output current range:||0-5.100A|
|Product weight:||66 grams|
|Output voltage setting and reading resolution:||0.01V|
|Output current setting and reading resolution:||0.001A|
|Measurement precision of output voltage:||+ (0.5%+1)|
|Output current measurement accuracy:||(1%+3 word)|
|Ripple (full load 40V to 32V/5A):||about 100mV peak.|
Note: we must ensure that the input voltage is more than 1.1 times the output voltage.
Digital tube display DP30V5A-L CNC power supply operation instructions:
1. open or close the output: under any state, the short ON/OFF key can open or close the output, circulate in turn, and when the output is closed, the digital tube has a "OFF" display indication.
2. set the output voltage and current: short press SET key, the digital tube starts to flicker, short press "+" button or "-" button to increase or reduce the output value, again short press SET key to switch "0.01V - >0.1V - >1V - >0.001A - >0.01A - >0.1A >1A - > adjustment", during the adjustment process, no operation automatically withdraws the keys in 6S. .
3. switching input and output display: under the condition of non output voltage and current setting, short pressing the "IN" button or "OUT" button can switch the display input voltage and output voltage. When the input voltage is displayed, the "IN" indicator lights up at the same time, indicating the output voltage, and the "IN" indicator is extinguished.
4.CC and CV indicate: the module operates under constant voltage or constant current state, and the CC and CV indicator lights will indicate the CC and CV state accordingly.
5. output current display zero point calibration: under no-load state, press and hold the SET key, do not put on power, carry out the output current display zero point calibration, calibration completion will automatically exit the calibration process, release the SET key. 6. set data automatic storage: in the process of key operation, the display state of the digital tube, the output voltage and current setting data, the switch output state automatically store memory, the next time the power is on the storage state.
Contact Person: Chengquan